Welch Allyn assumes no responsibility for any injury to anyone, or for any illegal or improper use of the product, that may result from failure to use this product in accordance with the instructions, cautions, warnings, or statement of intended use published in this manual.
Welch Allyn: Country of Origin: United States: Application: Battery Cover Screw Driver: For Use With: For Microtymp® 2 Portable Tympanometric Instrument: Specifications #1, With Welch Allyn Logo: UNSPSC Code: 42143500. Welch Allyn assumes no responsibility for any injury to anyone, or for any illegal or improper use of the product, that may result from failure to use this product in accordance with the instructions, precautions, warnings, or statement of intended use published in this manual. Page 3: Table Of Contents.
A cost-effective monitor with all the parameters for conscious sedation procedures.
a year ago
|a year agouser manual|
I would like a copy of the user manual for Welch Allyn 6200 Cardiac Monitor.
a year ago
|a year agoSerial Communication Issue|
Anyone have an info on the serial communication setup for the WA 6200? I'm using Windows 10, PuTTY and a RJ45 to USB cable to communicate with the unit. I can establish a serial connection and get the output from the unit but don't seem to be able to make any changes thru the serial port.Not sure if i need to type some special code or have PuTTY set up wrong..Reply
7 years ago
|7 years agoWA 6200 CO2 calibration gas needed|
I am trying to find a cal gas for this unit: 10% Adobe cs6 for mac download.
CO2; 10% O2; Balance N2. Air LiquideScott medical indicates that their T4724SRBD is the appropriate gas in 10 Liter aerosol cans: the cost is $131.00 each but, as you might expect, there is a 6 can minimum. Does anyone either have a single can they wish to part with or a desire to split this order??Reply